- Poster presentation
- Open Access
Evaluation of process parameters in shake flasks for mammalian cell culture
© Platas et al.; licensee BioMed Central Ltd. 2013
- Published: 4 December 2013
- Reynolds Number
- Specific Growth Rate
- Power Input
- Shake Flask
- Shake Flask Cultivation
Shake flask cultivation is nowadays a routine technique during process development for mammalian cell lines. During shaken culture, changes in agitation velocity, shaking diameter or shake flask size affect the hydrodynamics in the shake flask. This might be reflected in the growth of the cultured cells.
Process parameters such as power input, mixing time, fluid velocity etc. have been determined and described mathematically for shake flasks used for microbial cultivation, but only to some extend for mammalian cell culture. Especially the relationship between these parameters and growth characteristics of mammalian cells is still a relatively uncovered issue.
In this work, process parameters like specific power input, mixing time, maximum fluid velocity and Reynolds number were determined for four different shake flasks (baffled and unbaffled) in a range of shaking velocities on a shaking machine. The specific growth rate (μmax) of the human industrial cell line AGE1.HN® (ProBioGen AG, Berlin, Germany) was compared to the respective process parameters.
Power input ( P/V ) was calculated according to experimental data, that have been published in correlations with the form of Np = f(Re), where Np is the power number and Re the Reynolds number of the culture. The first correlation is based on the work by Büchs et al. [1, 2], who used a modified Np analog to bioreactors, and fited the experimental Np' data to Re. The second correlation used is based on the work of Kato et al. . Here, the calculation of the Reynolds number considers the diameter of the shaker (d o ) instead of the inner flask diameter (d i ).
Mixing time (Θ 95 ) was determined by means of the decolourization method (I/KI titrated with Na2S2O3). Decolourization time course was video recorded and visually analyzed.
Maximum fluid velocity ( u i ) was calculated at the maximum flask's inner diameter.
Reynolds number (Re) was calculated as Re = ρNd 2 /η, with d = d i , and d = d o , for the methods published by Büchs et al., and Kato el al. respectively.
A modified di (d i,mod ) was used for calculations of parameters in baffled flasks. This number considers the flask's depth into the flask circumference. The average specific growth rate μ max was employed as indicator for growth performance.
Our results point to regions of the studied parameters, where common operation windows can be identified for μmax. In these process windows the cells show a similar μmax in different shake flask, making cell growth comparable. These process windows are common for the flasks, independently of their size and the number of baffles.
The data obtained in this work can be used for process standardization and comparability of results obtained in shaken systems i.e. to guarantee consistency of results generated during laboratory studies with mammalian cells.
This work is a part of SysLogics (FKZ 0315275A): Systems biology of cell culture for biologics, a project founded by the German Ministry for Education and Research (BMBF).
- Büchs J, Maier U, Milbradt C, Zoels B: Power consumption in shaking flasks on rotary shaking machines: I. Power consumption measurement in unbaffled flasks at low liquid viscosity. Biotechnol Bioeng. 2000, 68: 589-593.View ArticlePubMedGoogle Scholar
- Büchs J, Maier U, Milbradt C, Zoels B: Power consumption in shaking flasks on rotary shaking machines: II. Nondimensional description of specific power consumption and flow regimes in unbaffled flasks at elevated liquid viscosity. Biotechnol Bioeng. 2000, 68: 594-601.View ArticlePubMedGoogle Scholar
- Kato Y, Hiraoka S, Tada Y, Shirai S, Koh ST, Yamaguchi T: Power-consumption of horizontally shaking vessel with circulating motion. Kagaku Kogaku Ronbunshu. 1995, 21: 365-371.View ArticleGoogle Scholar
- Platas O, Jandt U, Phan LDM, Villanueva ME, Schaletzky M, Rath A, Freund S, Reichl U, Skerhutt E, Scholz S, Noll T, Sandig V, Pörtner R, Zeng AP: Evaluation of criteria for bioreactor comparison and operation standardization for mammalian cell culture. Eng Life Sci. 2012, 12: 518-528.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.